Rossmax International Ltd., 12F - 189 Kang Chien Road, Taipei 114, TAIWAN ROC.
Blood pressure, Atrial fibrillation
Primary Client Use:
Intended for self-measurement and home use
Single measurements only
Discontinued but still available
Availability according to Countries or Regions:
The Rossmax X5 (V1) is an automatic blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate. Blood pressure measurements are taken from the upper arm. It is intended for self-measurement and home use. This device has been discontinued by the manufacturer but may be available through certain outlets.
While the technology used in the Rossmax X5 (V1), to measure blood pressure, has been declared as being equivalent to that used in a clinically validated device, no evidence has been published to show that the devices have been compared according to a protocol compliant with (EU) 2017/745 and MEDDEV 2.7/1 rev 4.
||Although equivalence to another device is claimed, this has not been tested to MDR requirements.
||BIHS (UK and IRL)
||Manufacturer declaration of equivalence
||The BIHS recommendation, despite the absence of the scientific evidence required by the same authors' criteria.
, Alvita CG-155FB
, Aspen C5B
, Aspen CF155FB
, Lloyds CardioZ
, Rossmax AC1000f (V1)B
, Rossmax CF155fB
, Rossmax CF175f (V1)B,P
, Rossmax CH155f (V2)B
, Rossmax X1 (V1)B,P
, Rossmax X3 (V1)B,P
, Rossmax X5 (V1)B
, Rossmax X9B,P
, Rossmax Accumed Carrdio C5B
, Rossmax X5 (V2)P
, Rossmax CH155f (V1)B
, Rossmax X9 BTZ
, Rossmax Z1B,P
, Rossmax Z5B,P
Legend: B BIHS Derivative
, P Stride Equivalence
, Z Almost Certain
Validation Publications for Equivalent Devices:
Note: This is a provisional list, as equivalence according to EU Regulation 2017/745 (e.g. MEDDEV 2.7/1 rev 4) is not proven. Accordingly, these publications are not used in the assessment of star-ratings.
Rossmax CF175f (V1)
Zhang L, Kang YY, Zeng WF, Li Y, Wang JG. Validation of the Rossmax CF175 upper-arm blood pressure monitor for home blood pressure monitoring according to the European Society of Hypertension International Protocol revision 2010. Blood Press Monit. 2015 Apr;20(2):104-7. doi: 10.1097/MBP.0000000000000089. PMID: 25350783.
ESH-IP:2010 - Pass General population