|BP||Medaval||None||Although equivalence to another device is claimed, this has not been tested to MDR requirements.|
|BP||DHL (DE)||General use (2019)||Manufacturer declaration of equivalence|
|BP||Stride BP||Self-measurement for adults and pregnancy (Preferred)||Equivalence claim without published evidence|
Legend: B BIHS Derivative, D DET Equivalence, P Stride Equivalence, Y Internal declaration
Note: This is a provisional list, as equivalence according to EU Regulation 2017/745 (e.g. MEDDEV 2.7/1 rev 4) is not proven. Accordingly, these publications are not used in the assessment of star-ratings.
Omron BP760N (HEM-7320-Z)
Takahashi H, Yoshika M, Yokoi T. Validation of two automatic devices for the self-measurement of blood pressure according to the ANSI/AAMI/ISO81060-2:2009 guidelines: the Omron BP765 (HEM-7311-ZSA) and the Omron BP760N (HEM-7320-Z). Vasc Health Risk Manag. 2015 Jan;9(11):49-53. doi: 10.2147/VHRM.S72438. PMID: 25657587.
81060-2:2009 - Pass General population
Grover-Páez F, Cardona-Muñoz EG, Cardona-Müller D, Guzmán-Saldívar VH, Rodríguez-De la Cerda M, Jiménez-Cázarez MB, Totsuka-Sutto SE, Alanis-Sánchez GA, Ramos-Becerra CG. Validation of the Omron HEM-7320-LA, upper arm blood pressure monitor with Intelli Wrap Technology Cuff HEM-FL1 for self-measurement and clinic use according to the European Society of Hypertension International Protocol revision 2010 in the Mexican population. Blood Press Monit. 2017 Dec;22(6):375-8. Epub: 2017 Sep 22. doi: 10.1097/MBP.0000000000000290. PMID: 28945665 .
ESH-IP:2010 General population (Note: Major violation in DBP range and distribution)
With HEM-FL1 Cuff
Omron M3 Comfort (HEM-7134-E)
Topouchian J, Hakobyan Z, Asmar J, Gurgenian S, Zelveian P, Asmar R. Clinical accuracy of the Omron M3 Comfort® and the Omron Evolv® for self-blood pressure measurements in pregnancy and pre-eclampsia - validation according to the Universal Standard Protocol. Vasc Health Risk Manag. 2018 Aug 31;14:189-97. doi: 10.2147/VHRM.S165524. PMID: 30214220. Available from: PMC6124447. eCollection 2018.
81060-2:2013 - Pass Normotensive pregnancy, Hypertensive pregnancy and Pre-eclampsia (n=45) (Note: Only one set of ISO results provided.)
BHS:1993 - Pass (A/A) Normotensive pregnancy, Hypertensive pregnancy and Pre-eclampsia (n=45)
Validation under the "Universal Protocol" is claimed, though only a consensus document was written, at that time, as referenced. Analysis appears to be according to Criterion 1 of ISO 81060-2:2013 and to BHS 1993.
Takahashi H. Validation of Home Blood Pressure-monitoring Devices Omron EVOLV (HEM-7600T-E), HEM-9210T, and M3 Comfort (HEM-7134-E) According to European Society of Hypertension International Protocol (ESH-IP) Revision 2010. J Clin Physiol (Jpn). 2019 Feb 1;49(1):37-46. doi: 10.34363/jocp.49.1_37. Available from: www.jstage.jst.go.jp.
ESH-IP:2010 - Pass General population
Omron M6 Comfort IT (HEM-7322U-E)
Chahine MN, Topouchian J, Zelveian P, Hakobyan Z, Melkonyan A, Azaki A, Diab R, Harb A, Asmar R. Validation of BP devices QardioArm® in the general population and Omron M6 Comfort® in type II diabetic patients according to the European Society of Hypertension International Protocol (ESH-IP). Med Devices (Auckl). 2017 Dec 27;11:11-20. doi: 10.2147/MDER.S142126. PMID: 29343992. Available from: PMC5749560. eCollection 2018.
ESH-IP:2010 - Pass Type 2 Diabetes Mellitus