Nihon-Kohden Vismo PVM-4000
Distributer: Nihon Kohden UK Ltd., Unit 118 Trident Court, 1 Oakcroft Road Chessington, Surrey KT9 1BD, UNITED KINGDOM.
Other: Nihon Kohden Corporation, 1-31-4 Nishiochiai, Shinjuku-ku, Tokyo 161-8560, JAPAN.
Blood pressure, Arrhythmias, SpO2, Temperature, ECG, Respiration
Primary Client Use:
Intended for patient monitoring
BP & SpO2: Finger
BP & SpO2: Continuous measurement
Availability according to Countries or Regions:
The Nihon-Kohden Vismo PVM-4000 is a patient monitor. Its blood pressure measurement technology has been proven to be accurate and information regarding the accuracy of its oxygen saturation measurement technology does not appear to be available. Blood pressure measurements and oxygen saturation measurements are taken from the finger. It is intended for bedside patient monitoring.
The technology used in the Nihon-Kohden Vismo PVM-4000, to measure blood pressure, has been compared to a clinically validated device, but the results of the equivalence study have not been published and cannot be verified; nor can the protocol be checked for compliance with (EU) 2017/745 and MEDDEV 2.7/1 rev 4. There appears to be no peer-reviewed clinical validation information available on the technology used in the Nihon-Kohden Vismo PVM-4000 to measure peripheral oxygen saturation.
||Although equivalence to another device is claimed, this has not been tested to MDR requirements.
||Japanese Society of Hypertension
||Professional use (2021-2022)
||Implied equivalence to the Nihon-Kohden Vismo PVM-2701
Nihon-Kohden Vismo PVM-2701J
, Nihon-Kohden BSM-1700J
, Nihon-Kohden BSM-3000J
, Nihon-Kohden BSM-6000J
, Nihon-Kohden CSM-1500J
, Nihon-Kohden CSM-1700J
, Nihon-Kohden CSM-1900J
, Nihon-Kohden PVM-2700J
, Nihon-Kohden TEC-8000J
, Nihon-Kohden ZM-540PGJ
, Nihon-Kohden ZM-940PGJ
, Nihon-Kohden GZ-140PJ
, Nihon-Kohden PVM-9900J
, Nihon-Kohden RMC-5000J
, Nihon-Kohden SVM-7000J
, Nihon-Kohden TEC-5000J
, Nihon-Kohden ZS-640J
, Nihon-Kohden ZS-940PGJ
, Nihon-Kohden Vismo PVM-4000J
, Nihon-Kohden Cardiolife EMS EMS-1052J
Legend: J JHS Implied Equivalence
Validation Publications for Equivalent Devices:
Note: This is a provisional list, as equivalence according to EU Regulation 2017/745 (e.g. MEDDEV 2.7/1 rev 4) is not proven. Accordingly, these publications are not used in the assessment of star-ratings.
Nihon-Kohden Vismo PVM-2701
Alpert BS. Validation of the Nihon Kohden PVM-2701/Impulse-1 automated device by both AAMI (2002) and ISO standards testing. Blood Press Monit. 2012 Oct;17(5):207-9. doi: 10.1097/MBP.0b013e328359c48e. PMID: 22968109.
81060-2:2009 - Pass Adults (n=69) and children (n=41)
SP10:2002 - Pass Adults (n=69) and children (n=41)