|BP||Medaval||★★★★★ Certified Accuracy||Recent clinical validation; recent protocol with certified adherence; multiple clinical validations|
|BP||BIHS (UK and IRL)||Self-measurement||Published evidence|
|BP||Stride BP||Self-measurement for adults||Published evidence|
|BP||MDR Criteria||Self-measurement||Published evidence|
Legend: P Stride Equivalence
Bing S, Chen K, Hou H, Zhang W, Li L, Wei J, Shu C, Wan Y. Validation of the Microlife BP A200 Comfort and W2 Slim automated blood pressure monitors in a general adult population according to the European Society of Hypertension and the ANSI/AAMI/ISO 81060-2: 2013 protocols. Blood Press Monit. 2016 Apr;21(2):118-23. Epub: 2015 Dec. doi: 10.1097/MBP.0000000000000169. PMID: 26683381.
ESH-IP:2010 - Pass General population
81060-2:2013 - Pass General population
Medaval Certification Assessment
Medaval Ltd. Accreditation assessment of the blood pressure measurement technology used in the Microlife W2 Slim wrist monitor, as validated according to the European Society of Hypertension International Protocol revision 2010 and also the AAMI/ANSI/ISO 81060-2:2013 standard for a general study in adults. Medical Device Assessment. 2016 Aug 5;2016(1604) 8 p. Epub: 2019 Jan 31. Available from: www.medaval.ie.
With proven validity of validation results according to an accepted protocol, the clinical accuracy of the blood pressure measurement technology, used in the Microlife BP W2 Slim, is certified by Medaval Ltd.